Potato Cassava Starch Machine,Cassava Starch Processing Machine,Potato Starch Extract Machine,Cassava Starch Making Machine Shuangfeng Nongjiabao Machinery Technology Co., Ltd , http://www.starchmakingmachine.com
I. Main Features 1. High intelligence: The use of single-chip microprocessors to complete the range of automatic conversion, temperature compensation and instrument self-test, there is no function switch and adjustment knob.
2. Multi-parameter display on the same screen: Simultaneous display of conductivity and temperature on one screen. The use of high-brightness LED display module, the main display with red 0.56-inch digital display conductivity value, eye-catching, visible distance; Deputy display with red 0.5-inch digital display temperature. To meet the user's different usage habits.
3. Industrial Control Watchdog: Make sure the meter does not crash.
II. Operation principle of the instrument Measurement principle: In order to avoid electrode polarization, the instrument generates a square wave signal with high stability and is added to the conductivity cell. The current flowing through the conductivity cell is proportional to the conductivity of the solution to be measured. After the high-impedance op amp is converted into a voltage, a programmable signal is amplified, detected and filtered to obtain a potential signal reflecting the conductivity; the microprocessor alternately samples the temperature signal and the conductivity signal, and after the operation and the temperature compensation, the measured solution is obtained. Conductivity at 25Â°C and current temperature.
Principle of temperature compensation: The conductivity of the electrolyte solution is affected by temperature changes and must be temperature compensated. In general, the temperature coefficient of a weak aqueous solution is 2 c/oÂ°C.
III. How to Use the Instrument The new meter must be set to â€œElectrode Constantâ€ and â€œTemperature Constantâ€ when used for the first time. The electrode constant is marked on the electrode. The temperature coefficient is set to 2.00% when shipped from the factory.
1. Function key description Key function electrode constant Press this key to quickly set and view the electrode constant temperature coefficient Press this key to set and view the temperature compensation coefficient OK Press this key to accept the setting Scroll (â–¼, â–²) can be used to change the labor in the measurement mode Temperature setting and changing the electrode constant and temperature compensation coefficient measurement Press this key to return to the measurement mode. When the instrument is powered on, the meter model and electrode constant value are displayed first. After 5 seconds, the instrument automatically enters the measurement state. H-1 is displayed.
The electrode constant indicates that the tester is in set mode and it is used to set or view the cell constants.
The temperature compensation coefficient indicates that the tester is in setting mode. It is used to set or view the temperature compensation coefficient.
ATC (automatic temperature compensation)
Indicates that the automatic temperature compensation probe is connected and operational. Manual temperature compensation will be activated when the ATC signal is not displayed.
Â°C indicates the temperature value.
Î¼S/cm, unit of measurement in ms/cm.
E-21 indicates the over-measurement range.
2. Parameter setting electrode constant setting Press the electrode constant key, the electrode constant light, the main display electrode constant value, press â–², â–¼ keys to increase or decrease the value, hold and hold, the speed of change will be faster and faster, to The required electrode constant value, press the enter key to accept the electrode constant setting, the electrode constant value flashes, the electrode constant will automatically return to the measurement state after the light goes out.
When setting the electrode constant, you can quickly change the electrode constant value by pressing the electrode constant key. Each time you press, the electrode constant is changed between 0.0100, 0.100, 1.000, and 10.00.
Note: The electrode constants set in the meter must be consistent with the constants marked on the electrodes used!
Temperature coefficient setting Press the temperature coefficient button, the temperature coefficient light is on, the main display shows the temperature coefficient value, press â–², â–¼ keys to increase or decrease the value, hold it down, the speed of change is faster and faster, to the desired temperature coefficient value Press the ENTER key to accept the temperature coefficient setting, the temperature coefficient value will flash, and the temperature coefficient will automatically return to the measurement state after the lamp is turned off. We suggest that when the conductivity of the water sample is less than 1Î¼S/cm, the temperature coefficient is 4.55%; when the conductivity is less than 10Î¼S/cm, the temperature coefficient is selected to be 4.00%; when the conductivity is less than 100Î¼S/cm, the temperature coefficient is selected to be 2.10%; higher conductivity The rate is set based on the temperature coefficient of the main ions in the water.
Note: If you only want to view the temperature coefficient value without modification, press the temperature compensation factor key to view, and press the measurement key to directly bring back the measurement status.
The temperature is manually set to facilitate the user's needs in a specific situation, the temperature of the watch can be manually set in the range of 0 ~ 60 Â°C. Once the manual temperature is set, the ATC light will go out.
In the manual warm-up state, you can use the â–², â–¼ keys to increase or decrease the value, hold it down, the speed of change will be faster and faster. Can be set to the desired temperature value.
Potato Cassava Starch Machine,Cassava Starch Processing Machine,Potato Starch Extract Machine,Cassava Starch Making Machine
Shuangfeng Nongjiabao Machinery Technology Co., Ltd , http://www.starchmakingmachine.com